FAQ: “What is the limit on the standardization in TTRAN. Using the small sample flask, filled with Ultra pure water. If they follow the protocol, they expect to find values (CieLab: 100.0; 0.0; 0.0)?” Continue reading
FAQ: “Do we have to keep the white tile at the reflectance port during transmission readings?” Continue reading
How do I know what mode my sensor is currently standardized in and how can I change modes?
The easiest way to tell what mode the instrument is using in the EasyMatch QC software is to locate the status bar in EasyMatch QC. The status bar is located at the very bottom of the software indicated in red in the picture below. This provides information about the sensor in the current mode, and when the standardization for that mode expires.